MICROELECTRONICS. 2N/91/ 2N/94/ COMPLEMENTARY SILICON POWER TRANSISTORS. 2N, 2N, 2N AND 2N ARE . Characteristics of the 2N bipolar transistor. Type – p-n-p; Collector-Emitter Voltage: 40 V; Collector-Base Voltage: 40 V; Emitter-Base Voltage: 5 V. Manufacturer Part No: 2N Order Code: Technical Datasheet: 2N Datasheet. Availability: No Longer Manufactured. See all Technical Docs.
||23 July 2008
|PDF File Size:
|ePub File Size:
||Free* [*Free Regsitration Required]
The soldering does not look original. Substituting a 2NG for the B hasn’t made any difference.
2N Datasheet PDF – Motorola
The photo titles show the channel allocations to the rails. Will check my box of scraps. Hopefully this will help with interpretation of my earlier msg.
So the problem seems to lie on the power supply board. Hardware shot simply shows the Motorola circled red.
Seems that’s the way it should be. As far as I can tell U and Q are behaving as expected but I can find no bad passives on the -5V supply sub-circuit. The new topic will begin with this message. I’ve started to compare how the -9V and -5V supplies are behaving as dahasheet first appears to be working fine and the second is not and they’re not very dissimilar subcircuits and use the same op-amp.
I can post some annotated scans of the relevant schematic parts if that would help. One of the LeCroy shots has parameters overlaid showing min and max voltages reached – as does the single Rigol shot taken simultaneously. Your email is included with the abuse report. Also can anyone make use of some fast switching diodes and other parts?
In this case the device is connected to a large pad on the The Group moderators are responsible for maintaining their community and can address these issues.
I hope that will shed some more light. And closer inspection shows you’re right that it’s vatasheet B At present, I’m just looking for advice on whether this part of the supply is particularly sensitive to the characteristics of Q, details of what the full identifier for the Motorola B might be and where I might find a datasheet, and advice as to whether the 2NG is a suitable replacement and, if not, what to use instead.
Are you sure you wish to delete this message from the message archives of TekScopes groups. I have plenty of photos and waveforms from diagnostics so far but I’m not sure how to upload them or link them to this post.
CENTR 2N Series Datasheets. 2N, 2N, 2N Datasheet.
Reason Report to Moderators Catasheet think this message isn’t appropriate for our Group. This cannot be undone. Biggest challenge is finding enough time to do the diagnostics.
Datasheey Mon, Oct 8, at Use a mica washer when mounting on a metal surface. Subject of the new topic: I’ve been unable to find any datasheet on this item and don’t know whether it’s original or a substitute from an earlier rework. If you had a spare lying around it would be good to have one available.
My eyes are not what they used to be. Presume there’s guidance on the site somewhere – will try some more searching. Search Only Within This Topic. I’ll bet the B in B is an 8 and if so is a date code. Reed Dickinson toggle quoted message. There’s also one expanded timebase shot from the LeCroy showing the U op-amp output into the base of Q and its correspondence with movements on the -5V rail, which starts its downward trajectory but is derailed by the positive shift at Q base and doesn’t recover.
Q’s Tek part number for my ‘s serial number isfor which Sphere lists 2N as a substitute. I’ll presume for now the 2N is working OK and move on.