Lab supply | First online laboratory shop, and selling where new features were introduced. practice. Instrumental factor must be checked. Chema Diagnostica will not be responsible for instrumental bad programming. Mindray BAA. Mindray) owns the intellectual property rights to this Mindray product and this manual. This manual may refer to information protected by copyrights or patents.
||25 May 2007
|PDF File Size:
|ePub File Size:
||Free* [*Free Regsitration Required]
Page 30 After completing the serial port setting, click Enable Para. After unplugging the adapters, pinch the buckles on the pump head and detach the pump shell. To wipe off dust from the system nindray, use a soft, clean and wet not too wet cloth, soaked with mild soap solution if necessary, to clean the surface.
Install pump head on the pump, with motor shaft into installation hole Pinch buckles to lock the pump head After installing the pump head, connect the tubing with the tubing that goes through the backboard and the waste tubing. To print the manual completely, please, download it. It will take mindrya minutes to upgrade the operating software. Page 29 surface of the lamp base.
Unique Diagnostics » Semi Automated BA 88A
Page 32 Figure Flow cell assembly structure 3. RSC Hard disk the disk on which the software is installed: BAA semi-auto chemical analyzer for clinic.
The configuration procedures are shown as follows: Semi Auto Analyser in Bengaluru. Driver Of The Step Motor 4.
Small laboratories; Pet hospitals Features: During configuration, modifying parameters other than temperature parameters should not be allowed. View Contact Call Seller Now.
And Fluid and then select Filter wheel home position to enter the following screen. By pressing ‘print’ button you will print only current page.
ba-88a semi-auto chemistry analyzer
Pull out ba-88q tubing that goes through the backboard of the analyzer for mm until the adapter is exposed. Photometer assembly includes lamp assembly, lamp base assembly, flow cell assembly and other components, which are used to position and install the optical components, thermal components and analogue board.
Wash the tubing with distilled water or DI water before shutting down the analyzer for instrument installed with flow cell. ,indray necessary, you can disconnect the connection of some cables in advance. Please pay attention to the following: The system is an analyzer designed for in vitro quantitative determination of clinical chemistries in serum, plasma, urine and CSF samples.
Enter text from picture: We have carved a niche amongst the most trusted names in this business, engaged in offering a comprehensive range of Bio Chemistry Analyzer A The procedure for adjusting is shown as follows: Click Lower upgr and the Lower computer upgrading dialog box will be displayed.
External keyboard and mouse support via Mnidray 5. Besides a TFT mmindray screen of 7 inches, it has multi-language operation software that is used with ease. If obvious blow-up is found, skewed installation of the lens might be the cause and the readjustment might be needed.
After completing the serial port setting, click Enable Para. Contact For Free Shipping. After calibration is completed, calibration formula can not be gained, but the previous calibration formula can be used to calculate the formula and the test will not be affected. Figure Lamp assembly structure WARNING If you want to replace the lamp, first switch off the mindtay and then wait at least 15 minutes for the lamp to cool down before touching it.