The HP B is a single channel W DC programmable power supply with GPIB and RS connectivity. It is part of the HP x series. Agilent B/B/B/B System DC Power. Supplies All references to HP-IB have been changed to GPIB. A note about. I just picked up two HP B power supplies as I read somewhere quickly that it not only can be programmed via GPIB to set voltage/current.
||25 August 2012
|PDF File Size:
|ePub File Size:
||Free* [*Free Regsitration Required]
Get fast answers from reviewers. Set the output current to its full scale value see Table B If you want to quickly view 66632b content of pages found on the following pages of the manual, you can use them. Using the online preview, you can quickly view the contents and go to the page where you will find the solution to your problem with HP Hewlett-Packard B.
Unscrew the line np cap from the rear panel and verify that the fuse rating matches what isspecified on the FUSES label on the rear panel. Selftest ErrorsPressing the Shift, Error keys will show the error number.
Page 51 Specifications – ATable A I also used a a, although the manual calls for an 8. ASpecificationsSpecificationsTable A-1 lists the specifications of the dc source.
Keysight Technologies (Agilent HP) B used or new for sale at used-line
It still seems to be in cal from 5 years ago. On the Function keypad, press Protect. Many people prefer to read the documents not on the screen, but in the printed version. Our goal is to provide you with a quick access to the content of the user manual for HP Hewlett-Packard Ho.
Current Measurement Low Range Action7. Page 22 3 – InstallationLocationThe outline diagram in figure gives the 6362b of your dc source. For verification tests, see appendix B.
HP (Hewlett-Packard) 6632B user manual
Press the Prot Clear key to clear theprotection condition. Enter the new password from the keypad.
Wait for the DMM readingto stabilize. CError MessagesError Number ListThis appendix gives the error numbers and descriptions that are returned by the dc source. Mail will not be published required. The dc sourceis shipped with its HP-IB address set to 5. 6632 56 B – Verification and Calibration5. CopyleftAll Wrongs 6632g. Enter the existing password from Entry keypad and press Enter3.
US and upHP B: Failure to comply with these precautions or with specific warnings elsewhere in this manual violates safetystandards of design, manufacture, and intended use of the instrument. Shopbop Designer Fashion Brands.
Jp the Function keypad, press Output. I got the service manual and it turns out you can defeat password protection by flipping a DIP switch inside the chassis.
The display returns to Meter mode when the calculation is complete. How you did the voltage calibration? The dc source must beinstalled in a location that allows sufficient space at the sides and back of the unit for adequate aircirculation see Bench Operation.
After much swearing, compressed air, and windex, the inside of the chassis looked like this: The following table lists the errors that are associated with SCPI syntax errors and interfaceproblems. Installation – 3Under normal operation, the dc source senses the output voltage at the output terminals on the backof the unit.
It also made the fan run quieter. After much swearing, compressed air, and windex, the inside of the chassis looked like this:. Page 50 A – SpecificationsSupplemental CharacteristicsTable A-2 lists the supplemental characteristics, which are not warranted but are descriptions oftypical performance determined either by design or type testing.