Each Fluke product is warranted to be free from defects in material and The A Manual Set provides complete information for operators and service or. Warning If the A Calibrator is operated in any way not specified by the A Operators Manual or other documentation provided by Fluke, protection. View and Download FLUKE A operator’s manual online. Multi-Product Calibrator. A Test Equipment pdf manual download.
||18 August 2005
|PDF File Size:
|ePub File Size:
||Free* [*Free Regsitration Required]
Entering Scope Cal mode prior to having the SC enabled for at least 5 minutes will cause a warning message to be displayed. Testing Harmonics Volts Performance Then set the sampling heads to the settings listed below, to establish a reference signal.
Turn the A Calibrator power on. High Frequency Flatness Verification at flke. Page 74 A Operator Manual Actual revision numbers replace 1. Refer to “Checking A Status” for more information.
To print the manual completely, please, download it. Refer to Figure to identify the spur.
A Operator Manual
Table lists the interface messages that the A Calibrator accepts. Adi listing Short-Term Edge. Calibration The standard A has no internal hardware adjustments. General specifications that apply to the Calibrator Mainframe hereafter termed the Calibrator can be found in Chapter 1. Oscilloscope Input Capacitance Measurement Specifications. Page The minimum use specifications are usually determined by maintaining a specified test uncertainty ratio between the calibration equipment and the unit under test.
See Figure for connections, and see Table for shunt information. Each assembly is accompanied by an illustration showing the location of each part and its reference designator. DCV oltage V erific ation.
The frequency range of human hearing; normally 15 – 20, Hz. The HPA digital multimeter can be programmed from either the front panel or over the remote interface to make these measurements.
If the temperature return for the internal reference is 0 0. The Calibrator is operated under the conditions specified in Chapter 1. They are not the calibration constants obtained when the A was calibrated by the factory before shipment.
Adjust A90R13 to set the edge signal occurring between 0 ns and 2 ns to the reference point set above.
Features of the A Calibrator include the following: Once set, the A Calibrator retains the temperature sensor type until power off or reset. Sets the output of the A and establishes a new reference point for the error mode.
Repeat steps 4 and 5 for all 5500s frequencies listed in Table Page A Operator Manual Table Wave Generator Amplitude Verification: Description Text characters 0 to Allow the A reading to stabilize, then enter the reading into Column A of the table. It is unlikely the fuses will blow unless there is another fault since the regulators will current limit manua, the fuse ratings.
General Specifications Warmup Time Twice the time since last warmed up, to a maximum of 30 minutes.
Fluke a – Sc Multi-product Calibrator | eBay
Vertical deflection characteristics are verified by calibrating the voltage gain. For the phase test, ac couple the input to the phase meter. The Oscilloscope Calibration Option is described in the Options chapter.
For example, if testing a K thermocouple, the cable and miniconnector are for a K thermocouple.
Fluke 5500A/COIL Calculator User Manual
D Apply power sensor correction factor for 10 MHz W: Maintenance Cleaning the Air Filter To clean the air filter, refer to Figure and proceed as follows: Watts uncertainty power voltage Setup options can be selected using the softkeys under the control display. Remove the power switch pushrod. Setting Voltage And Current Limits When the calibration process is compete, you are prompted to either store all the correction factors in nonvolatile memory or discard them and start over.
Press to reconnect the signal output. Connections for Calibrating DC Current. Exploded View of Rear Panel Assemblies. Then follow these steps to calibrate pulse width: Leveled Sine Wave flatness is calibrated at multiple amplitudes.
Accessing the Fuse and Selecting Line Voltage Calibrating the Voltage Amplitude on an Oscilloscope The oscilloscope voltage vertical gain is calibrated by applying a dc or low frequency square wave signal and adjusting its gain to meet the height specified for different voltage levels, as designated by the graticule line divisions on the oscilloscope.
Adjusting the Leveled Sine Wave Function There is one adjustment procedure that needs to be made for the leveled sine wave function. Summary of Commands and Queries This section describes commands and queries that are used specifically for the oscilloscope calibration option. If you are, the Control Display shows the following menu.